This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||1 July 2007|
|PDF File Size:||13.1 Mb|
|ePub File Size:||19.98 Mb|
|Price:||Free* [*Free Regsitration Required]|
To ensure that the protection provided by this equipment is not impaired, do not use or sc1100 this equipment in any manner other than that specified in this manual.
Hach-Lange SC 100 User Manual
Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure. Don’t show me this message again. Troubleshooting Table 11 presents sensor warnings displayed in the Hahc Log, possible causes, and corrective actions. Section 1 Specifications Section 1 Specifications Specifications are subject to change without notice.
The data logs store the measurement data at selected intervals. Calibrate Sensor as required by regulatory agency Per agency-dictated schedule. Troubleshooting Section 7 Troubleshooting Replacing The Lamp Assembly Twist the bulb in a counterclockwise direction, pulling out slightly, until it is released from the housing.
In the United States: Rinse with deionized water before calibrating. Table of Contents Section 1 Specifications Wire gauge less than 18 AWG is not recommended. Dry Verification System Startup Two types of verifications are offered.
Page 50 Maintenance a. Maximum cable length is 9. Index Sample Line Installation Measured sc10 based on a gain of 1. See Replacement Parts and Accessories on page 55 for ordering information. The controller contains three relays designed for use with high voltage Danger: Page 23 Allow at least mm Troubleshooting Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions.
Table of Contents Add to my manuals Add. Refer to Figure 29 and perform the following steps to replace the fuses: The frequency will depend on the nature and concentration of dissolved and suspended solids in the sample.
Don’t have an account? Changing The Sensor Name Operation 4.
System Startup Two types of verifications are hacu. Page 39 Operation 4. Printed in the U. Theory Of Operation General Information 2. Close and secure the cover. In addition, the passcode also regulates function selection for relay options.
Page of 72 Go. The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk.
Hach-Lange SC User Manual | 58 pages
The turbidimeter sensor must be mounted within six feet of the controller unless an extension cable is used. To print the manual completely, please, download it. Instrument Setup Operation 4. Table of Contents Section 5 System Startup Index Numerics 90 Degree Detector Page 35 Operation 4.
The display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed to a different value. Section 10 Repair Sx100 Authorization must be obtained from Hach Company before sending any items for repair.
The calibration history log shows the gain value, the time and date of the calibration, and the initials of the operator performing verification.
Restore sample flow to the instrument. Section 1 Specifications Specifications are subject to change without notice. The time is available only in hour military format. Disconnect and remove the existing wires between the quick connect and terminal block J5, see Figure 21 on page 4.
Controller Display Features Operation 4. Section 7 Troubleshooting 7. To hold the output until released: Remove the quick connect fitting and wires and install the threaded sc1000 on the opening to maintain the environmental rating. Controller Dimension Illustrations Installation 3.
Hach sc100 1720E Instrument Manual
Failure to do so could result in serious injury to the operator or damage to the equipment. Wear cotton gloves to protect your hands and to avoid fingerprints on the bulb. Setting Up System Security Operation 4.